Regarding applicability to a range of HPLC flow rates, the signal level of analytes by APPI has been observed to saturate and even decay at higher solvent flow rates (above 200 μl/min), and therefore, much lower flow rates are recommended for APPI than for ESI and APCI. This has been suggested to be due to absorption of photons by the increasing density of solvent molecules., However, this leads to the benefit that APPI can extend to very low flow rates (e.g., 1 μL/min domain) allowing for effective use with capillary LC and capillary-electrophoresis.