The Displayed Average Noise Level (DANL) is just what it says it is—the average noise level displayed on the analyzer. This can either be with a specific resolution bandwidth (e.g. −120 dBm @1 kHz RBW), or normalized to 1 Hz (usually in dBm/Hz) e.g. −150 dBm(Hz).This is also called the sensitivity of the spectrum analyzer. If a signal level equal to the average noise level is fed there will be a 3 dB display. To increase the sensitivity of the spectrum analyzer a preamplifier with lower noise figure may be connected at the input of the spectrum analyzer.