Following Rollins' discovery in 1920 that lead aprons protected against X-rays, lead aprons with a lead thickness of 0.5 mm were introduced. Due to their weight, lead-free and lead-reduced aprons were subsequently developed. In 2005, it was recognized that in some cases the protection was significantly less than wearing lead aprons. The lead-free aprons contain tin, antimony and barium, which have the property of producing intense radiation (X-ray fluorescence radiation) when irradiated. In Germany, the Radiology Standards Committee has taken up the issue and introduced a German standard (DIN 6857-1) in 2009. The international standard IEC 61331-3:2014 was finally published in 2014. Protective aprons that do not comply with DIN 6857-1 of 2009 or the new IEC 61331-1 of 2014 may result in higher exposures. There are two classes of lead equivalency classes: 0.25 mm and 0.35 mm. The manufacturer must specify the area weight in kg/m at which the protective effect of a pure lead apron of 0.25 or 0.35 mm Pb is achieved. The protective effect of an apron shall be appropriate to the energy range used, up to 110 kV for low energy aprons and up to 150 kV for high energy aprons.