Thermometers calibrated per ITS–90 use complex mathematical formulas to interpolate between its defined points. ITS–90 specifies rigorous control over variables to ensure reproducibility from lab to lab. For instance, the small effect that atmospheric pressure has upon the various melting points is compensated for (an effect that typically amounts to no more than half a millikelvin across the different altitudes and barometric pressures likely to be encountered). The standard even compensates for the pressure effect due to how deeply the temperature probe is immersed into the sample. ITS–90 also draws a distinction between "freezing" and "melting" points. The distinction depends on whether heat is going into (melting) or out of (freezing) the sample when the measurement is made. Only gallium is measured while melting, all the other metals are measured while the samples are freezing.