Heating in diamond-anvil cells is typically done by two means, external or internal heating. External heating is defined as heating the anvils and would include a number of resistive heaters that are placed around the diamonds or around the cell body. The complementary method does not change the temperature of the anvils and includes fine resistive heaters placed within the sample chamber and laser heating. The main advantage to resistive heating is the precise measurement of temperature with thermocouples, but the temperature range is limited by the properties of the diamond which will oxidize in air at 700 °C The use of an inert atmosphere can extend this range above 1000 °C. A tungsten ring-wire resistive heater inside a BX90 DAC filled with Ar gas was reported to reach 1400 °C. With laser heating the sample can reach temperature above 5000 °C, but the minimum temperature that can be measured when using a laser-heating system is ~1200 °C and the measurement is much less precise. Advances in resistive heating are closing the gap between the two techniques so that systems can be studied from room temperature to beyond 5700 °C with the combination of the two.