The figure shows the main components of an APPI source: a nebulizer probe which can be heated to 350–500 °C, an ionization region with a VUV photon source, and an ion-transfer region under intermediate pressure that introduces ions into the MS analyzer. The analyte(s) in solution from the HPLC flows into the nebulizer at a flow rate that can range from μL/min to mL/min range. The liquid flow is vaporized by nebulization and heat. The vaporized sample then enters into the radiation zone of the VUV source. Sample ions then enter into the MS interface region, frequently a capillary through the combination of a decreasing pressure gradient and electric fields.