Additional effects on tube rigidity will cause shifts in the calibration factor over time due to degradation of the flow tubes. These effects include pitting, cracking, coating, erosion or corrosion. It is not possible to compensate for these changes dynamically, but efforts to monitor the effects may be made through regular meter calibration or verification checks. If a change is deemed to have occurred, but is considered to be acceptable, the offset may be added to the existing calibration factor to ensure continued accurate measurement.