Potentiometric type dvm, Electrical Engineering

Potentiometric type DVM: A potentiometric type of DVM, employs voltage comparison technique. In this DVM the unknown Voltage is compared with a reference voltage value is fixed by the setting of the calibrated potentiometer. The potentiometer setting is changed to obtain balance (i.e. null condition).When null conditions are obtained the value of the unknown voltage, is indicated by the dial setting of the potentiometer.

In potentiometric type DVMs, the balance is not obtained manually but is arrived at automatically. Thus, this DVM is in fact a self-balancing potentiometer .The potentiometric DVM is provided with a readout which displays the voltage being measured.

The block diagram of basic circuit of a potentiometric DVM is shown in.

The unknown voltage is filtered and attenuated to suitable level. This input voltage is applied to a comparator (also known as error detector). This error detector may be chopper. The reference voltage is obtained from a fixed voltage source. This L voltage is applied to a potentiometer R. The value of the feedback voltage depends upon the position of the sliding contact; the feedback voltage is also applied to the comparator.

The unknown voltage and the feedback voltages are compared in the comparator. The output voltage of the comparator is the difference of the above two voltages. The difference of voltage is called the error signal. The error signal is amplified and is fed to a potentiometer adjustment device which moves the sliding contact of the potentiometer. This magnitude by which the sliding contact moves depends upon the magnitude of the error signal. The direction of movement of slider depends upon whether the feedback voltage is larger or the input voltage is larger. The sliding contact moves such a place where the feedback voltage equals the unknown voltage. In that case, there will not be any error voltage and hence here will be no input to the device adjusting the position of the sliding contact and therefore it (sliding contact) will come to rest. The position of the potentiometer adjustment device at this point is indicated in numerical form on the digital readout device associated with it. Since the position at which no voltage appears at potentiometer adjustment device is the one where the unknown voltage equals the feedback voltage, the reading of readout device indicates the value of unknown voltage. The reference voltage source must be extremely stable and generally consists of a standard cell or a Zener diode sources.


Posted Date: 7/12/2012 6:12:33 AM | Location : United States

Related Discussions:- Potentiometric type dvm, Assignment Help, Ask Question on Potentiometric type dvm, Get Answer, Expert's Help, Potentiometric type dvm Discussions

Write discussion on Potentiometric type dvm
Your posts are moderated
Related Questions

Calculate the diameter of copper wire of length 100 metres used as winding material in a transformer such that the resistance of the whole winding is 2 ohms. Calculate the diameter

Q. Discuss the nature of sinusoidal signal? Discuss the nature of evenness and oddness of: (a) The sinusoidal signal x(t) = A cos(2πf 0 t + θ). (b) The complex exponentia

In use, an appropriate bias voltage is applied between the two bases, with B2 made positive with respect to B1. Because the N-type bar is resistive, a relatively small current will

If a high resolution emcoder is used for position control, will the number of pole count in the DC motor be matter? What is the requirement for the DC motor in order to work with h

up-down counter

CURRENTLY, orthogonal frequency division multiplexing (OFDM) is emerging as the preferred modulation technique in modern high data rate broadband wireless mobile communication syst

Traffic Light Display Design a three color LED system with the following times for each LED: Duration of Green signal : 16s Duration of Yellow signal : 1s Duration of R


Is it used to replace the basic gates ?