﻿ Use of microwater meter-武汉中科新仪电力科技极速PK拾


Use of microwater meter

1. Connect the SF6 equipment

Connect the thread end of the measuring line with the switch joint, tighten the screw with the wrench, and close the needle valve on the other end of the measuring line.

Insert the quick joint end of the test line into the sampling port on the micro water meter。

Connect the exhaust pipe to the air outlet.

Finally, connect the switch joint to the SF6 electrical equipment measurement interface, and tighten with the wrench;

2. Check the battery

This instrument is recommended to use alternating current。

When using direct current, please check the battery power shown in the upper right corner. If the battery is lower than about 20%, please switch off and use after charging.

3. Start measuring

First all open the instrument to measure the needle valve on the pipe, then use the flow valve on the panel to adjust the flow, adjust the flow rate to about 0。5 L/M, start to measure SF6 dew point。

The first device takes 5 to 10 minutes, and each device takes 3 to 5 minutes。

4. Stored data

After the equipment measurement is completed, the data can be saved in the instrument, and the operation menu can be adjusted by "ok", and the specific operation mode can be seen in the next section.

5. Measure other equipment

After measuring a device, close the needle valve on the gauge line and the regulator on the microwater meter. Remove the adapter from the SF6 electrical equipment. If you need to continue to measure other devices, follow the above steps to continue to measure the next device.

6. End of measurement

After all equipment measurements are completed, close the micro-water instrument source.

广西快3走势 新疆喜乐彩走势图 安徽快3计划 山东11选5 一分时时彩 728彩票计划群 韩国1.5分彩 秒速快3 荣鼎娱乐 新疆喜乐彩app