{"action":"created","issue":{"url":"https://api.github.com/repos/PaxInstruments/t400-firmware/issues/160","labels_url":"https://api.github.com/repos/PaxInstruments/t400-firmware/issues/160/labels{/name}","comments_url":"https://api.github.com/repos/PaxInstruments/t400-firmware/issues/160/comments","events_url":"https://api.github.com/repos/PaxInstruments/t400-firmware/issues/160/events","html_url":"https://github.com/PaxInstruments/t400-firmware/issues/160","id":58509382,"number":160,"title":"Temperatures below 17.6 C","user":{"login":"charlespax","id":281642,"avatar_url":"https://avatars.githubusercontent.com/u/281642?v=3","gravatar_id":"","url":"https://api.github.com/users/charlespax","html_url":"https://github.com/charlespax","followers_url":"https://api.github.com/users/charlespax/followers","following_url":"https://api.github.com/users/charlespax/following{/other_user}","gists_url":"https://api.github.com/users/charlespax/gists{/gist_id}","starred_url":"https://api.github.com/users/charlespax/starred{/owner}{/repo}","subscriptions_url":"https://api.github.com/users/charlespax/subscriptions","organizations_url":"https://api.github.com/users/charlespax/orgs","repos_url":"https://api.github.com/users/charlespax/repos","events_url":"https://api.github.com/users/charlespax/events{/privacy}","received_events_url":"https://api.github.com/users/charlespax/received_events","type":"User","site_admin":false},"labels":[],"state":"open","locked":false,"assignee":null,"milestone":null,"comments":1,"created_at":"2015-02-22T16:59:23Z","updated_at":"2015-02-22T17:36:03Z","closed_at":null,"body":"Below temperatures of approximately 17.6 C the thermocouples appear as if they are disconnected.\r\n\r\nBelow is an image of what happened when I put thermocouple channel 2 in a refrigerator. Channel 1 had been in the refrigerator for some time. Channel 3 is in the air. Channel 4 is not connected.\r\n\r\n![img_20150223_001729](https://cloud.githubusercontent.com/assets/281642/6319282/75758e58-baf1-11e4-8616-5ec148105600.jpg)"},"comment":{"url":"https://api.github.com/repos/PaxInstruments/t400-firmware/issues/comments/75446848","html_url":"https://github.com/PaxInstruments/t400-firmware/issues/160#issuecomment-75446848","issue_url":"https://api.github.com/repos/PaxInstruments/t400-firmware/issues/160","id":75446848,"user":{"login":"charlespax","id":281642,"avatar_url":"https://avatars.githubusercontent.com/u/281642?v=3","gravatar_id":"","url":"https://api.github.com/users/charlespax","html_url":"https://github.com/charlespax","followers_url":"https://api.github.com/users/charlespax/followers","following_url":"https://api.github.com/users/charlespax/following{/other_user}","gists_url":"https://api.github.com/users/charlespax/gists{/gist_id}","starred_url":"https://api.github.com/users/charlespax/starred{/owner}{/repo}","subscriptions_url":"https://api.github.com/users/charlespax/subscriptions","organizations_url":"https://api.github.com/users/charlespax/orgs","repos_url":"https://api.github.com/users/charlespax/repos","events_url":"https://api.github.com/users/charlespax/events{/privacy}","received_events_url":"https://api.github.com/users/charlespax/received_events","type":"User","site_admin":false},"created_at":"2015-02-22T17:36:03Z","updated_at":"2015-02-22T17:36:03Z","body":"Oh I see. This is happening because of\r\n```\r\n  if(microVolts > maxConversion || microVolts < -200){\r\n    return OUT_OF_RANGE;\r\n  }\r\n```\r\n\r\nWhen I comment out that code I am able to measure lower temperatures."}}